I REMOTE CONTROL MANUAL g g. II. II. II. | g. II g. II g. I li g a. Serial Number. May MODELS /14/20/24/30/ DUAL- AND QUAD-CHANNEL. 5 Manual Operation. Vertical Controls. Time-Base Controls. Probes. Bandwidth Limit. Trigger controls (Standard). Displaying Traces. Display Control and. See Remote Control Manual. 6FRSH ([SORUHU. Also via GPIB and RSC, the Scope Explorer software integrates the oscilloscope with Windows 95 and.
||23 September 2006
|PDF File Size:
|ePub File Size:
||Free* [*Free Regsitration Required]
All processing occurs in function memories E and F which may be displayed on the screen by simply pressing the appropriate function button. Processing is fully auto- matic and is simultaneous whenever more than one function has been selected.
Full text of “LeCroy Service Manual”
The timing of subsequent triggers in sequence mode is measured with 0. Turn off all the traces, except CHI. They may not be the problem but at some point they will be. Wave- forms and measurements are presented instantly on the screen and can be transferred directly to a wide variety of plotters and printers, archived onto PCs, or stored on an optional credit card memory facility.
Teledyne LeCroy – Product Manuals – Oscilloscopes
It will remain high for a read cycle. I didn’t know that Teledyne acquired Lecroy! My fingers are crossed that this cap went bad and it’s the only damage. Set trigger source to EXT.
Cheers, Gil Wow that was very kind of you! Multi-source trigger operational modes: This cannot be undone. The 8 intensity bits are converted to analog current by an 8-bit DAC.
These features are combined with genuine 1 and 2 millivolt sensitivity settings and a big, crisp display to let you see the finest signal structure. I also do this for every board you plan to work on. In operation, the current drawn from the 11 kV supply anode current may be up to uA.
Reference and function memories: The display processor waits until the user enables it to access the memory by sending the first frame synchro. In parallel a 1 KOhm resistor is added. Window Functions Rectangular, von Hann Hanning.
Four different window functions are applied to indicate their effect on leakage suppression and spectral resolution.
Okay, so I’ve now noticed the manual they sent me was for theexactly the same one in the archive linked from here!
So I need to save up all my spare pennies and buy myself a very expensive SM2. Users can obtain time and fre- quency values simultaneously and compare phases of the various frequency components with each other.
By effectively removing high- frequency noise, with digital smoothing functions, waveforms can be analyzed with resolution from 8 to 11 bits. Waveform operations can be performed on live, stored, processed or expanded waveforms. Last not least comprehensive protection circuitry is implemented. This is the fastest access: They are selected through simple menus that allow functions to be chained together allowing more complex computations.
Do not place the solder side of the board directly on an antistatic foam or mat, which are slightly conducting and can discharge the battery. Relative weighting factors can be chosen from 1: The data and address busses are 16n bits wide.
If lecroh some reason you don’t get it, please post here again. It appears to have been custom made for Agilent as it is not a standard Manal series part.
So I just need lecroh get it working. If one of these three lines is low, SHUT is low.
LeCroy 9410 Manuals
Highest side lobe indicates the reduction in leakage of signal components into neighboring frequency bins. This represents the major use of the display board.
Set the triggers Smart Trigger: The least significiant lecryo represents 0. Cheers, Gil toggle quoted message. To remove it, first remove the top and bottom covers, and the power supply support In this way the envelope of all waveforms is accumulated for up to a maximum of 1 , sweeps.
So it’s actually not as good. The line WE is set low during a write cycle to enable the memory data load. LeCroy is the registered trademark of LeCroy Corporation. It is easy to get confused where things go when you have a bunch of chips removed from their sockets! I may be less happy when I fix the blown cap and it’s still broken, but I’ll still be quite happy.
The ZC register, internal to the MDS, is not reloaded with a new value until the vector in progress is completed i.
I’ve checked the voltages from the PSU not under load and they areall correct.