Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
||20 December 2012
|PDF File Size:
|ePub File Size:
||Free* [*Free Regsitration Required]
IIRC, the sweep parameters were something like this: Message 5 of 6. Message 6 of 6. Message 4 of 6.
HP A for Sale|Impedance Analyzers|Impedance Measurement Products|Test Equipment Center
Good luck with it Most Active Software Boards: Hp419a1 2 of 6. Should I be looking at doing it with LabView or something else? However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at. It’s 84MB, so I hope you have a fast link. If not, what would creating a driver for it entail?
I am able hp4119a communicate with the device through the GPIB interface, and am also able to retrieve information.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: Guess you never know until manuap try! I have not worked with that instrument, but the Agilent web site has the operating and service manual here.
Message 3 of 6.
Has anyone done anything on interfacing with the A since Matt C’s last post? I think the would probably take the old commands, but the ALSO has new commands.
I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different. Matt – you’re right, the commands for the A are completely different from the – probably because of its age. Accepted by topic author Matt C.
The forums emoticons took over for a bit there. In the code above, MHz is an arbitrary frequency. You really need the manual.
Message 1 of 6. I suspect that this difference might be causing my problems. In particular, after setting up and initiating the instrument, I am initiating majual measurement and reading it as follows:.
I have worked with the A and, as I recall, the commands are nothing like any other instrument I have seen.
Agilent / HP 4191A RF Impedance Analyzer
Sorry, that last line of code there is supposed to read: When I hp191a it a try, though, it worked fine. Thanks for your help guys.
Would you mind posting the code you used on the A?